Wash your hands with warm soapy water and dry thoroughly. Prepare the lancing device.
Check the use by date on the test strip container. Do not use test strips past the use by date.
Remove a test strip from the test strip container. Close the cap tightly.
Insert the metallic end of the test strip into the meter. The meter turns on.
When a flashing drop symbol appears, perform a fingerstick with the lancing device.
Gently squeeze your finger to assist the blood flow. This helps you get a blood drop.
Touch the yellow edge of the test strip to the blood drop. Remove your finger from the test strip when the flashing hourglass symbol appears. Do not put blood on top of the test strip.
Your test result appears with an arrow. This arrow shows if your test result falls above, within, or below the target range.* The target range is represented by the green region of the target range indicator. The arrow will flash if your test result falls above or below this range.
Remove and discard the used test strip.
Reference 1. ISO 15197:2013. 2. Accu-Chek Instant package insert. 3. Breitenbeck N, et al., J Diabetes Sci Techno 2017, 11(4):854-855. 4. Kocher S., et al., J Diabetes Sci Technol 2009;3(5):1136-1143.
Disclaimer: Accu-Chek Softclix lancing device is for single patient use only. The same device must not be used for multiple patients
The Bluetooth® word mark and logo are registered trademarks and are owned by the Bluetooth SIG, Inc.
From 0 customers
Only logged in customers who have purchased this product may leave a review.
The Accu-Chek Instant is connected meter with Bluetooth technology, synchronise effortlessly with mySugr app. The Accu‑Chek Instant meter makes testing effortless. The features which are not heavily used have been removed from the meter and placed on the Accu‑Chek Connect online. This makes handling and interactions with the system effortless. Additionally, the Accu‑Chek Instant meter has an intuitive target range indicator that instantly shows Person with diabetes how they are doing with their management.
The Accu‑Chek Instant meter requires two batteries to power the added value features, such as the backlit display and Bluetooth connectivity. These features are not available with the Accu‑Chek Instant S version, so only one battery is required.
As a general rule, with the connectivity of the meter enabled, the battery life is anticipated to be at least 750 tests. Normal testing without connectivity should produce approximately 1,000 test results for both systems.
Flagging of test results is not possible on the meter itself. However, if a patient chooses to do so, they can add this information via mySugr app and the Accu‑Chek Connect account.
Test results may be seen in 7, 30 and 90-day averages.
At least 720 readings can be stored inside the meter. However only the last result and the 7, 30 and 90-day averages can be reviewed on the meter. To review all values the meter must be connected to one of the software solutions.
The IDF Guidelines are used for the High values range and the ADA Guidelines for the Low values range. In range: 70-160 mg/dL, below range: 69-10 mg/dL, above range: 161-600 mg/dL.
Yes, the target ranges in the meter can be changed using the device settings on the Accu‑Chek Connect Online system.
10/10 references an ISO-compliant internal testing protocol. For three different lots of test strips, 95% of BG results must fall within ±10 mg/dL of reference result at glucose concentrations < 100 mg/dL or within ± 10% at glucose concentrations ≥ 100 mg/dL.
The Accu‑Chek Instant meter is one of the most accurate systems our organization has ever manufactured and meets ISO 15197:2013/EN ISO 15197:2015 standards for accuracy. It has 10/10 analytical accuracy.
No, the Accu‑Chek Instant meters are designed to ensure that the battery power level will not impact results given by the meter.
The Accu‑Chek Instant meters is our latest development in accuracy from a pioneer in blood glucose testing technologies. This is one of the most accurate meters in our portfolio.
No, Accu‑Chek Instant meters do not require manual coding.
No, there is no compatibility between the Accu‑Chek Instant test strips with these and other Accu‑Chek systems. Accu‑Chek Instant test strips are compatible with Accu‑Chek Instant S Meter.
The Accu‑Chek Instant test strip requires only a small speck of blood (0.6 μL).
As with today’s products, the user will receive an E-4 error code.
These cookies are necessary for the website to function and can not be turned off in our systems. They are usually set only in response to actions taken by you, e.g. a request for services such as your personal preference settings, logging in or filling out a form. You can set your browser to block or warn you about these cookies. Parts of the website may therefore not work. These cookies do not store personal data.
These cookies allow for improved functionality and personalization. They may be set by us or by third party providers whose services we have added to our pages. If you do not enable these cookies, some or all of these functionalities may not work properly.
These cookies allow us to count visitors and traffic sources so that we can measure and improve the performance of our website. They help us find out which pages are most and least popular, as well as how visitors move around the website. We may pass these cookies on to companies that help us run these analytics tools. All information that these cookies collect is aggregated and can not be directly related to you. If you choose not to allow these cookies, we will not receive information about when you visited our website, and thus could not measure the website's effectiveness.
These cookies are set via our site by our advertising partners. They can be used by these companies to build a profile of your interests and show you relevant advertisements on other websites. They do not store your direct personal data, but identify your unique browser and device. If you do not allow these cookies, you will experience less targeted advertising across different websites.