1qty Accu-Chek Instant S Glucometer, 1qty Accu-Chek Softclix Lancing Device, 10 qty Accu-Chek Softclix Lancet, 1 qty Pack of 10 Accu-Chek Active 10 Test Strips, 1 Qty Carry Case, User Manuals, and Quick Start Guide
Authorised Seller : Hasmukh Agency
Country of Origin: China
11 Penetration Depth
0.8 To 2.3 Mm
Precise Lancing, Less Risk Of Skin Damage
The Intuitive target range indicator gives you visual reassurance. It can be individualized to suit your personal therapy goals*
Only what you need – Your test result and your averages (7/30/90 days) are visible on the meter
No set up required – Just insert a strip, apply a small blood sample, and read your blood glucose result on the large display
Easy-edge test strip – Apply a small drop of blood anywhere along the wide, yellow edge
Proven accuracy guaranteed by the makers of the Accu-Chek brand, fulfilling the requirements of the ISO 15197:2013^
Test strips come in a 10, 25, 50 vial, so you can choose whatever fits your budget or therapy
Enjoy virtually pain-free testing with the Accu-Chek Softclix lancing device
* Blood glucose values (Lows≤69 mg/dL, Normal 70-160 mg/dL, High≥161/dL) are pre-set on the Target Range Indicator and must be interpreted in conjunction with the appropriate medical condition at the time of taking reading, and in accordance with advice of your physician.
^ Criterion A of clause 6.3 for the lot tested (Freckmann et al., JDST, 6(5), 2012)
Wash your hands with warm soapy water and dry thoroughly. Prepare the lancing device.
Check the use by date on the test strip container. Do not use test strips past the use by date.
Remove a test strip from the test strip container. Close the cap tightly.
Insert the metallic end of the test strip into the meter. The meter turns on.
When a flashing drop symbol appears, perform a fingerstick with the lancing device.
Gently squeeze your finger to assist the blood flow. This helps you get a blood drop.
Touch the yellow edge of the test strip to the blood drop. Remove your finger from the test strip when the flashing hourglass symbol appears. Do not put blood on top of the test strip.
Your test result appears with an arrow. This arrow shows if your test result falls above, within, or below the target range.* The target range is represented by the green region of the target range indicator. The arrow will flash if your test result falls above or below this range.
Remove and discard the used test strip.
Reference 1. ISO 15197:2013. 2. Accu-Chek Instant S package insert. 3. Breitenbeck N, et al., J Diabetes Sci Techno 2017, 11(4):854-855. 4. Kocher S., et al., J Diabetes Sci Technol 2009;3(5):1136-1143
From 0 customers
Only logged in customers who have purchased this product may leave a review.
Yes, the Accu‑Chek Instant S meters comply with the requirements of ISO 15197:2013/EN ISO 15197:2015 (In vitro diagnostic test systems – Requirements for blood glucose monitoring systems for self-testing in managing diabetes mellitus).
No, the Accu‑Chek Instant S meters are designed to ensure that the battery power level will not impact results given by the meter.
The Accu‑Chek Instant S system fulfils the ISO 15197:2013/EN ISO 15197:2015 standards and delivers even tighter 10/10 accuracy for reliable results you can trust.
10 references an ISO-compliant internal testing protocol. For three different lots of test strips, 95% of BG results must fall within ±10 mg/dL of reference result at glucose concentrations < 100 mg/dL or within ± 10% at glucose concentrations ≥ 100 mg/dL 10/
As with today’s products, the user will receive an E-4 error code.
No. There is no ability to squeeze extra blood from the finger to dose a strip a second time. However, there is a slight pause in testing that allows for the patient to take full use of the available blood from their finger.
No, there is no compatibility between the Accu‑Chek Instant S test strips with these and other Accu‑Chek systems
No, Accu‑Chek Instant S meters do not require manual coding.
At least 720 readings can be stored inside the meter. However only the last result and the 7, 30 and 90-day averages can be reviewed on the meter. To review all values the meter must be connected to one of the software solutions.
Flagging of test results is not possible on the meter itself. However, if a patient chooses to do so, they can add this information via Accu‑Chek Connect Online account.
These cookies are necessary for the website to function and can not be turned off in our systems. They are usually set only in response to actions taken by you, e.g. a request for services such as your personal preference settings, logging in or filling out a form. You can set your browser to block or warn you about these cookies. Parts of the website may therefore not work. These cookies do not store personal data.
These cookies allow for improved functionality and personalization. They may be set by us or by third party providers whose services we have added to our pages. If you do not enable these cookies, some or all of these functionalities may not work properly.
These cookies allow us to count visitors and traffic sources so that we can measure and improve the performance of our website. They help us find out which pages are most and least popular, as well as how visitors move around the website. We may pass these cookies on to companies that help us run these analytics tools. All information that these cookies collect is aggregated and can not be directly related to you. If you choose not to allow these cookies, we will not receive information about when you visited our website, and thus could not measure the website's effectiveness.
These cookies are set via our site by our advertising partners. They can be used by these companies to build a profile of your interests and show you relevant advertisements on other websites. They do not store your direct personal data, but identify your unique browser and device. If you do not allow these cookies, you will experience less targeted advertising across different websites.